|Book Series (77)||
|Mechanical and process engineering||736|
|Mining and metallurgy||29|
|Architecture and civil engineering||58|
|ISBN-13 (Hard Copy)||9783736971462|
|Lamination of Cover||matt|
|Place of Dissertation||Braunschweig|
Common electrical engineering
|Keywords||Non-linear fiber optics, stimulated Brillouin scattering, fiber optic sensors, polymer optical fiber, humidity, strain, temperature, CYTOP, PFGI-POF, multimode fiber, graded-index fiber, coupling, BOFDA, distributed sensing, distributed Brillouin sensors, optical filtering, frequency uncertainty, propagation of light, Nicht-lineare Faseroptik, stimulierte Brillouin-Streuung, faseroptische Sensoren, optische Polymerfaser, Feuchtigkeit, Temperatur, Belastung, Multimode-Faser, Gradientenindex-Faser, verteilte Abtastung, verteilte Brillouin-Sensoren, optische Filterung, Frequenzunsicherheit, Ausbreitung des Lichts|
In this thesis, a distributed Brillouin sensor in perfluorinated polymer optical fibers utilizing BOFDA is presented. These commercially available polymer fibers offer beneficial characteristics for sensing applications such as higher break down strain up to 100 %, minimal bending radii below 2 mm, higher sensitivity to temperature and lower sensitivity to strain compared to their silica equivalent.
The SBS parameters – backscattering power, linewidth and frequency shift – are inclusively related to the environmental parameters humidity, temperature and tensile strain. Furthermore, the perfluorinated polymer optical fibers are characterized with respect to the influence of relative humidity and temperature changes on spectral transmission absorption and Rayleigh backscattering, respectively.
The chosen wavelength of operation at 1319 nm corresponds to lower fiber propagation loss (< 37 dB/km) compared to other approaches at 1550 nm (150 – 250 dB/km). A 86 m PFGI-POF was successfully measured by BOFDA with spatial resolution of 3.4 m. Thus, the sensing range is demonstrated to be extended by a factor of four and the number of sensing points is increased by a factor of five. A non-strained fiber segment is successfully detected within a strained sensing fiber.